View a manual of the Medisana BU 514 below. All manuals on ManualsCat.com can be viewed completely free of charge. By using the 'Select a language' button, you can choose the language of the manual you want to view.
GB WARNING Read the instruction manual carefully before using this device, especially the safety instructions, and keep the instruction manual for future use. Should you give this device to another person, it is vital that you also pass on these instructions for use. Safety Instructions • The unit is intended only for use in the home. Consult your doctor prior to using the blood pressure monitor if you are concerned about health matters. Only use the unit for its intended purpose in accordance with the instructions for use. Warranty claims become void if the unit is misused. i Please consult your doctor prior to using the unit if you suffer from illnesses such as arterial occlusive disease. i The unit may not be used to check the heart rate of a pacemaker. i Pregnant women should take the necessary precautions and pay attention to their individual circumstances. Please consult your doctor if necessary. i Consult your physician to get more information about your blood pressure. A self-diagnosis and self-treatment based on the measured values may be dangerous. Always follow the instructions given by your physician! i If you experience discomfort during a measurement, such as pain in the upper arm or other complaints, press the START/STOP-button to release the air immediately from the cuff. Loosen the cuff and remove it from your arm. i The device only works correct with the ﬁtting cuff. i The unit is not suitable for children. i Do not allow children to use the unit. Medical products are not toys! i Please ensure that the unit is kept away from the reach of children. i Do not put the air tube around your neck - danger of suffocation! i The swallowing of small parts like packaging bag, battery, battery cover and so on may cause suffocation. i The user needs to check that the equipment functions properly and that it is in proper working condition before being used. i Only the cuff provided can be used, and cannot be changed to any other branded cuff. It can only be replaced with a cuff of exactly the same type. i The device may not be operated in rooms with high levels of radiation or in the vicinity of high-radiation devices, such as radio transmitters, mobile tele- phones or microwaves, as this could cause functional defects or incorrect measurements. i Do not use the equipment where ﬂammable gas (such as anaesthetic gas, oxygen or hydrogen) or ﬂammable liquids (such as alcohol) are present. i No modiﬁcation of this unit/device is allowed. i Please do not attempt to repair the unit yourself in the event of malfunctions, since this will void all warranty claims. Only have repairs carried out by authorised service centres. i Protect the unit against moisture. If liquid does manage to penetrate into the unit, remove the batteries immediately and do not continue to use the unit. In this case contact your specialist dealer or notify us directly. i Please do not use a dilution agent, alcohol or petrol to clean the unit. i Please do not hit the device heavily or let if fall down from a high place. i Do not use mobile phones in the range of the device. This may lead to mal- functions. i If the unit is not going to be used for a long period, please remove the batteries. • • • • • • • • • • • • • • • • • SAFETY NOTES FOR BATTERIES Do not disassemble batteries! i Replace all batteries if the unit display shows the low battery symbol. i Never leave any low battery in the battery compartment since it may leak and cause damage to the unit. i Increased risk of leakage! Avoid contact with skin, eyes and mucous mem- branes! If battery acid comes in contact with any of these parts, rinse the affected area with copious amounts of fresh water and seek medical attention immediately! i If a battery has been swallowed, seek medical attention immediately! i Replace all of the batteries simultaneously! i Only replace with batteries of the same type, never use different types of batteries together or used batteries with new ones! i Insert the batteries correctly, observing the polarity! i Remove the batteries from the device if it is not going to be used for an exten- ded period of 3 months or more! i Keep batteries out of children's reach! i Do not attempt to recharge batteries! There is a danger of explosion! i Do not short circuit! There is a danger of explosion! i Do not throw into a ﬁre! There is a danger of explosion! i Do not throw used batteries into the household refuse; put them in a hazardous waste container or take them to a battery collection point, at the shop where they were purchased! • • • • • • • • • • • • • • Blood pressure classiﬁcation WHO These values have been speciﬁed by the World Health Organisation (WHO) without making any allowance for age. Low blood pressure systolic <100 diastolic <60 Normal blood pressure (green display area ) systolic 100 - 139 diastolic 60 - 89 Forms of high blood pressure / hypertension Mild hypertension (yellow display area ) systolic 140 – 159 diastolic 90 – 99 Moderate hypertension (orange display area ) systolic 160 – 179 diastolic 100 – 109 Severe hypertension (red display area ) systolic ≥ 180 diastolic ≥ 110 Blood pressure that is too low represents just as great a health risk as blood pressure that is too high! Fits of dizziness may lead to dangerous situations arising (e.g. on stairs or in trafﬁc)! Inﬂuencing and evaluating readings • Measure your blood pressure several times, then record and compare the results. Do not draw any conclusions from a single reading. i Your blood pressure readings should always be evaluated by a doctor who is also familiar with your personal medical history. When using the unit regularly and recording the values for your doctor, you should visit the doctor from time to time to keep him updated. i When taking readings, remember that the daily values are inﬂuenced by se- veral factors. Smoking, consumption of alcohol, drugs and physical exertion inﬂuence the measured values in various ways. i Measure your blood pressure before meals. i Before taking readings, allow yourself at least 5-10 minutes rest. i If the systolic and diastolic readings seem unusual (too high or too low) on several occasions, despite correct use of the unit, please inform your doctor. This also applies to the rare occasions when an irregular or very weak pulse prevents you from taking readings. • • • • • Starting up Insert / replace batteries You must insert the batteries provided before you can use your unit. The lid of the battery compartment is located on the underside of the unit. Open it by slightly pushing it and moving it upwards. Remove it and insert the 4 AA LR6 type 1.5V batteries supplied. Ensure correct polarity when inserting the batteries (as marked inside the battery compartment). Close the battery compartment. Replace the batteries if the change battery symbol appears on the display or if nothing appears on the display after the unit has been switched on. 6 8 r 3. Month / date setting: The month ﬁgure starts to ﬂash for setting. Keep pressing the MEM button until the actual month ﬁgure appears. Press SET button to conﬁrm the month setting. Afterwards, the day needs to be set in the same way. After- wards, you can proceed with the time setting mode. 3 5 4. Time setting: The hour ﬁgure starts to ﬂash for setting. Keep pressing the MEM button i until the actual hour ﬁgure appears. Press SET button to conﬁrm the hour setting and proceed in the same way for minute setting. Conclusively, CL appears in the display. You may delete stored values at this point - please check chapter „Memory- clear of measurements“. Press SET button to leave the setting mode. If the batteries are changed, the set infor- mation is lost and needs to be input again. 5 Fitting the cuff 1. 2. 3. 4. 5. Push the end piece of the air tube into the hole on the left side of the unit prior to use. i Slide the open end of the cuff through the metal bracket so that the Velcro fastener is on the outside and it becomes a cylindrical form (Fig.1). Slide the cuff over your left upper arm. i Position the air hose in the middle of your arm in line with your middle ﬁnger (Fig.2) (a). The lower edge of the cuff should be 2 - 3 cm above the crease of the elbow. (b). Pull the cuff tight and close the Velcro fastener (c). i Measure the pulse on your bare arm. i Only position the cuff on the right arm if it cannot be used on the left arm. Al- ways carry out measurements on the same arm. i Correct measuring position for sitting (Fig.3). Discontinuing a measurement If it is necessary to interrupt a blood pressure measurement for any reason (eg. the patient feels unwell) the START/STOP button can be pressed at any time. The device immediately decrease the cuff pressure automatically. 4 Display stored values Symbol Cause Correction Error messages and error remedying Error indicators The following symbol will appear on the display when measuring abnormal: E-1 E-2 E-3 E-5 Weak signal or sudden pressure change External strong disturbance Error during inﬂation process Abnormal blood pressure Weak batteries Put on the cuff correctly. Repeat the measurement in the correct way. When near cell phone or other high radiant device, the measurement may fail. During measurement, do not move and do not speak. Put on the cuff correctly. Make sure, that the hose is properly connected. Repeat the measurement. Repeat the measurement after relaxing 30 minutes. If you get unusual readings for 3 times, please contact your doctor. The batteries are too low or empty. Replace all 4 batteries with new 1.5 V type AA LR6 batteries. • • The year ﬁgure starts to ﬂash for setting. Keep pressing the MEM button i until the actual year ﬁgure appears. Press SET button to conﬁrm the year setting. Afterwards, you can proceed with the month/date setting mode. 5 3 7 Disposal This product must not be disposed of together with domestic waste. All users are obliged to hand in all electrical or electronic devices, regardless of whether or not they contain toxic substances, at a municipal or commercial collection point so that they can be disposed of in an environmentally accep- table manner. Remove the batteries before disposing of the device/unit. Do not dispose of old batteries with your household waste, but at a battery collection station at a recyc- ling site or in a shop. Consult your local authority or your supplier for information about disposal. Explanation of symbols IMPORTANT Follow the instructions for use! Non-observance of these instructions can result in serious injury or damage to the device. WARNING These warning notes must be observed to prevent any injury to the user. CAUTION These notes must be observed to prevent any damage to the device. NOTE These notes give you useful additional information on the installation or operation. Device classiﬁcation: type BF applied part LOT number Manufacturer Date of manufacture Common factors of wrong measurements • Please keep quiet for 5-10 minutes and avoid eating, drinking alcohol, smoking, exercising and bathing before taking a measurement. All these factors will inﬂuence the measurement result. Remove any garment that ﬁts closely to your upper arm. Always measure on the same arm (normally left). Measurements should be taken regularly at the same time of each day, as the blood pressure varies even during the day. Any effort to support the arm during measurement may increase the measured blood pressure. Make sure, you are in a comfortable, relaxed posi- tion and do not move or constrict your muscles during measurement. Use a cushion to support your arm if necessary. If the arm artery lies lower or higher than the heart, a false reading will be obtained. A loose or open cuff causes false readings. With repeated measurements , blood accumulates in the arm which can lead to false reading. Consecutive blood pressure measurements should be repeated after 1 minute pause or after the arm has been held up in order to allow the accumulated blood to ﬂow away. • • • • • • • • 4 • • • • Scope of supply Please check ﬁrst of all that the unit is complete. The following parts are included as standard: • 1 MEDISANA blood pressure monitor BU 514 • 1 cuff with air tube • 4 batteries (type AA, LR6) 1.5 V • 1 storage pouch • 1 instruction manual Please contact your supplier if you notice any transport damage on unpacking the unit. What is blood pressure? Blood pressure is the pressure created in the blood vessels by each heart beat. When the heart contracts (= systole) and pumps blood into the arteries, it brings about a rise in pressure. The highest value is known as the systolic pressure and is the ﬁrst value to be measured. When the cardiac muscle relaxes to take in fresh blood, the pressure in the arteries also falls. Once the arteries are re- laxed, a second reading is taken – the diastolic pressure. How is blood pressure measured? The MEDISANA BU 514 is a blood pressure unit which is used to measure blood pressure at the upper arm. The measurement is carried out by a micro- processor, which, via a pressure sensor, measures the vibrations resulting from the inﬂation and deﬂation of the cuff over the artery. Settings 1. User setting: Press SET button when power off, or appear on the display. By pressing MEM-button you may choose between user , or . Press SET button to conﬁrm the user setting.Afterwards, you can proceed with the year setting mode. 5 3 5 The readings are automatically saved in the previously selected memory ( or ). Up to 90 measured values with date and time can be stored in each memory. The measurement readings remain on the display. The unit switches off automatically after approx. 180 seconds if no buttons are pressed. The unit may also be switched off using the START/STOPbutton . WARNING Do not take any therapeutic measures on the basis of a self measurement. Never alter the dose of a medicine prescribed by a doctor. 6. 7. 4 Please contact the distributor resp. the service centre if you can’t solve the problem. Do not disassemble the unit by yourself. Problem Check Cause and solutions No power No inﬂation Err appears and the measurement is aborted Leak cuff Troubleshooting Check the battery power and if the polarity is correct. Check for proper hose condition and if it is properly connected to the device. Check whether you have moved the arm during inﬂation or if you have talked during measurement. Check whether the cuff is too loose or damaged. Replace all batteries and pay attention to correct polarity during insertion. Connect the air hose properly. Use a new cuff. Keep your arm and your body still. Do not talk during measure- ments. Put on the cuff correctly and close it tightly. Use a new cuff. Cleaning and care Remove the batteries before cleaning the unit. Clean the unit using a soft cloth lightly moistened with a mild soapy solution. Never use abrasive cleaning agents, alcohol, naphta, thinner or gasoline etc.. Never immerse the unit or any component in water. Be cautiious not to get any moisture in the main unit. i Do not wet the cuff or attempt to clean the cuff with water. Using a dry cloth, gently wipe away any excess moisture that may remain on the cuff. Lay the cuff ﬂat in an unrolled position and allow the cuff to air dry. i Do not expose the unit to direct sunlight; protect it against dirt and moisture. Do not expose the unit to extreme hot or cold temperatures. Keep the device in the storage case when not in use. Store the unit in a clean and dry location. Guidelines / Standards This blood pressure monitor meets the requirements of the EU standard for non- invasive blood pressure monitors. It is certiﬁed in accordance with EC Guidelines and carries the CE symbol (conformity symbol) “CE 0297”. The blood pressure monitor corresponds to European standards EN 1060-3, EN 81060-1 and EN 81060-2. The speciﬁcations of EU Guideline “93/42/EEC of the Council Directive dated 14 June 1993 concerning medical devices” are met. You can request the complete Conformity Declaration from Medisana GmbH, Jagenbergstrasse 19, 41468 Neuss, Germany, or you can also download it from the Medisana homepage (www.medisana.de). Electromagnetic compatibility: (see attached leaﬂet) Name and model Display system Memory slots Measuring method Power supply Blood pressure measuring range Pulse measuring range Maximum error tolerance for static pressure Maximum error tolerance for pulse rate Pressure generation Deflation Automatic switch-off Operating conditions Storage conditions Dimensions (L x W x H) Cuff Weight (main unit) Item number EAN number Acessories MEDISANA Blood pressure monitor BU 514 Digital display 2 x 90 measurement values Oscillometric 6 V=, 4 x 1.5V batteries AA LR6 0 – 299 mmHg 40 – 199 beats/min. ± 3 mmHg ± 5 % of the value Automatic with pump Automatic After approx. 180 seconds +5 °C to +40 °C, 15 to 85 % max. relative humidity -20 °C to +55 °C, 10 to 85 % max. relative humidity Approx. 130 x 109 x 60 mm 22 - 36 cm for adults Approx. 333 g without batteries 51165 40 15588 51165 3 - Cuff M 22 - 36 cm for adults with average upper arm circumference Art. No. 51168 / EAN 40 15588 51168 4 - Cuff, L 32 - 42 cm for adults with upper arm of large circumference Art. No. 51169 / EAN 40 15588 51169 1 : : : : : : : : : : : : : : : : : : : : Technical specifications In accordance with our policy of continual product improvement, we reserve the right to make technical and visual changes without notice. The current version of this instruction manual can be found under www.medisana.com Please contact your supplier or the service centre in case of a claim under the warranty. If you have to return the unit, please enclose a copy of your receipt and state what the defect is. The following warranty terms apply: 1. The warranty period for MEDISANA products is 3 years from date of purchase. In case of a warranty claim, the date of purchase has to be proven by means of the sales receipt or invoice. 2. Defects in material or workmanship will be removed free of charge within the warranty period. 3. Repairs under warranty do not extend the warranty period either for the unit or for the replacement parts. 4. The following is excluded under the warranty: a. All damage which has arisen due to improper treatment, e.g. nonobservance of the user instructions. b. All damage which is due to repairs or tampering by the customer or unauthorised third parties. c. Damage which has arisen during transport from the manufacturer to the consumer or during transport to the service centre. d. Accessories which are subject to normal wear and tear as batteries, cuff etc.. 5. Liability for direct or indirect consequential losses caused by the unit are excluded even if the damage to the unit is accepted as a warranty claim. Warranty and repair terms For service, accessories and spare parts, please contact: MEDISANA GmbH, Jagenbergstr. 19, 41468 NEUSS, GERMANY. +44 20 38680977 email@example.com 3 5 i i i i After the cuff has been appropriately positioned,the measurement can begin: Taking a blood pressure measurement 1. 2. 3. 4. Switch the unit on by pressing the START/STOP button . i If the START/STOP button is pressed, all display characters are shown (display test). This test can be used to check that the display is indicating properly and in full. i The unit is ready for measurement. The number 0 ﬂashes for 2 seconds and two short beeps are emitted. The device then automatically inﬂates the cuff slowly in order to measure your blood pressure. The rising pressure in the cuff is shown on the display. i The unit inﬂates the cuff until sufﬁcient pressure has built up for a measure- ment. Then the unit slowly releases air from the cuff and carries out the measurement. When the device detects the signal, the heart symbol on the display starts to ﬂash, you can hear the beep for every heartbeat once the heartbeat signal is detected. 4 4 DE/GB 51165 09/2018 Ver.1.2 Instruction Manual Please read carefully! 1 2 2 3 4 5 6 7 8 9 q w e r t 0 z u i o Contraindications Proper use •The device is intended for measuring the blood pressure from the upper arm of an adult. •The device is not suitable for measuring the blood pressure of children. Ask your doctor before using it on older children. Persons who suffer from arrhythmia, diabetes, cardiovascular problems or who have had a stroke should consult their doctor before using the device. • IP21 Protection: IP21 GB Unit and LCD-Display Blood Pressure Monitor BU 514 Cover for Battery Compartment MEM button (Memory Recall) START/STOP button SET button Battery Compartment (on underside) Push-in Connector for air tube LCD screen (display) Display of Date/Time Display of Systolic Pressure Display of Diastolic Pressure Pulse symbol Display of Pulse Rate Change Battery symbol Arrhythmia display (arrhythmia cordis) Memory Location number Memory-Symbol Blood Pressure Indicator (green - yellow - orange - red) User memory 1 / 2 Unit and LCD display 1 2 4 6 5 7 8 9 q 0 w e r z t u i o 3 2. Year setting: Blood Pressure Monitor BU 514 This unit features 2 separate memories, each with a capacity of 90 memory slots. Results are automatically stored in the memory selected by the user. Press the MEM button when power off, to call up the measured values stored. Press the MEM button again All values average appear on the display. to call up the last measurement value. Repeatedly pressing the MEM button displays the respective values measured previously. If you have reached the ﬁnal entry and do not press a button, the unit automatically switches itself off after approximately 8 seconds in memory recall mode. Memory recall mode can be exited at any time by pressing the START/STOP button , which will also switch the unit off. The oldest value is deleted if 90 measured values have been stored in the memory and a new value is saved. 4 3 3 3
Have a question about the Medisana BU 514 but cannot find the answer in the user manual? Perhaps the users of ManualsCat.com can help you answer your question. By filling in the form below, your question will appear below the manual of the Medisana BU 514. Please make sure that you describe your difficulty with the Medisana BU 514 as precisely as you can. The more precies your question is, the higher the chances of quickly receiving an answer from another user. You will automatically be sent an e-mail to inform you when someone has reacted to your question.